TYPES OF BIOLOGICAL INDICATORS
There are at least three types of biological indicators. Each type of indicator incorporates a known species of a microorganism of known sterilization resistance to the sterilization mode. Some biological indicators may also contain two different species and concentrations of microorganisms.
One form of biological indicator includes spores that are added to a carrier (a disk or strip of filter paper, glass, plastic, or other materials) and packaged to maintain the integrity and viability of the inoculated carrier.
Carriers and primary packaging shall not contain any contamination (physical, chemical, or microbial) that would adversely affect the performance or the stability characteristics of the biological indicator. The carrier and primary packaging shall not be degraded by the specific sterilization process, which is used in a manner that will affect the performance of the biological indicator. The carrier should withstand transport in the primary and secondary packaging and handling at the point of use. The design of the carrier and primary packaging should minimize the loss of the original inoculum during transport, handling, and shelf life storage.
Another form of biological indicator is a spore suspension that is inoculated on or into representative units of the product to be sterilized. This represents an inoculated product; however, a simulated inoculated product may be used if it is not practical to inoculate the actual product. A simulated product is a preparation that differs in one or more ways from the actual product, but performs as the actual product using test conditions or during actual production sterilization processing. Spore suspensions with a known D value should be used to inoculate the actual or simulated product. If a simulated inoculated product is used, it must be demonstrated that it will not degrade the sterilization resistance of the bioindicator. The physical design of actual or simulated product can affect the resistance of spore suspensions that are inoculated on or into the products. In the case of liquid inoculated products, it is often advisable to determine both the D value and z value of the specific biological indicator microorganism in the specific liquid product. The population, D value, z value where applicable, and endpoint kill time of the inoculated actual or simulated product should be determined.
A third form of biological indicator is a self-contained indicator. A self-contained biological indicator is designed so that the primary package, intended for incubation following sterilization processing, contains the growth medium for recovery of the process-exposed microorganisms. This form of biological indicator together with the self-contained growth medium can be considered a system. In the case of self-contained biological indicators, the entire system provides resistance to the sterilization process.
If the biological indicator is a paper strip or disk in a self-contained package that includes an available culture medium, the package design should be readily penetrable by the sterilizing agent. To allow for the time lag that may occur while the sterilizing agent reaches the contained microorganisms in the system, the D value, process endpoint kill time, and the survival time should be characterized for the system and not solely for the paper strip in the self-contained unit. Following the sterilizing treatment, the spore strip or disk is immersed in the self-contained medium by manipulation, which allows contact with the culture medium.
Self-contained biological indicators may also consist of a spore suspension in its own medium, and they often also contain a dye, which indicates positive or negative growth following incubation. Resistance of the self-contained system is dependent upon penetration of the sterilant into the package. Penetration may be controlled by the manufacturer through varying designs and composition of the self-contained biological indicator package, ampul, or container. Self-contained ampul biological indicators may be incubated directly following exposure to the sterilization process. The entire system is then incubated under the specified conditions. Growth or no growth of the treated spores is determined visually (either by observing a specified color change of an indicator incorporated in the medium or by turbidity) or by microscopic examination of the inoculated medium.
The self-contained system resistance characteristics must also comply with the labeling of the self-contained system and the relevant biological indicator monograph. The self-contained biological indicator system should withstand transport in the secondary packaging and handling at the point of use without breakage. The design of the self-contained system should be such to minimize the loss of the original inoculum of microorganisms during transport and handling. During or after the sterilization process, the materials used in the self-contained system shall not retain or release any substance that can inhibit the growth of low numbers of surviving indicator microorganism under culture conditions. Adequate steps must be taken to demonstrate that the recovery medium has retained its growth support characteristics after exposure to the sterilization process.
All operations associated with the preparation of biological indicators are controlled by a documented quality system. Traceability is maintained for all materials and components incorporated in or coming into direct contact with the microorganism suspension, the inoculated carrier, or the biological indicator.
The preparation of stock spore suspensions of selected microorganisms used as biological indicators requires the development of appropriate procedures, including mass culturing, harvesting, purification, and maintenance of the spore suspensions. The stock suspension should contain predominantly dormant (nongerminating) spores that are held in a nonnutritive liquid.
The finished product (microbial suspension, inoculated carriers, or biological indicators) supplied by commercial manufacturers shall have no microorganisms, other than the test microorganism, present in sufficient numbers to adversely affect the product. The system to minimize the presence of microorganisms other than the biological indicator microorganism in the product will be validated, monitored, and recorded.
Selection for Specific Sterilization Processes
The selection of a biological indicator requires a knowledge of the resistance of the biological indicator system to the specific sterilization process. It must be established that the biological indicator system provides a challenge to the sterilization process that exceeds the challenge of the natural microbial burden in or on the product.
The effective use of biological indicators for the cycle development, process, and product validation, and routine production monitoring of a sterilization process requires a thorough knowledge of the product being sterilized, along with its component parts (materials and packaging). Only the widely recognized biological indicators specified in the particular biological indicator monograph should be used in the development or validation of a sterilization process. This will ensure that the biological indicator selected provides a greater challenge to the sterilization process than the bioburden in or on the product. Some users may require biological indicators with characteristics that differ from those widely available commercially. In such cases, users may grow their own spore cultures for the express purpose of preparing in-house biological indicators for their specific use. In such a case, the user is well advised to use organisms already described in the scientific literature as indicator organisms, and the user must have the capability of determining D and z values for in-house biological indicators. When biological indicators are prepared in-house, users must confirm the population, purity, and shelf life of the biological indicator to ensure the validity of any test conducted using the in-house biological indicator. When a bioburden-based sterilization process design is used, data comparing the resistance of the biological indicator to that of bioburden are essential. Enumeration of the bioburden content of the articles being sterilized is also required. The process must result in a biologically verified lethality sufficient to achieve a probability of obtaining a nonsterile unit that is less than one in a million.
Alternatively, the overkill method may be used in the design of a sterilization process. In this case, specific assumptions are made regarding the resistance assumption used in establishing sterilization process lethality requirements. In general, all overkill processes are built upon the assumption that the bioburden is equal to one million organisms and that the organisms are highly resistant. Thus, to achieve the required probability of a nonsterile unit that is less than one in a million, a minimum 12 D process is required. A 12 D process is defined as a process that provides a lethality sufficient to result in a 12 log reduction, which is equivalent to 12 times a D value
for organisms with sufficiently higher resistance than the mean resistance of bioburden. Because the bioburden is assumed to be one million, an overkill process will result in a probability of nonsterility at much less than 106
in actual practice. Overkill process design and evaluation may differ depending upon the sterilization process under test. The use of an overkill design and validation approach may minimize or obviate the need for bioburden enumeration and identification.
For moist heat sterilization process, spores of suitable strains of Bacillus stearothermophilus are commercially available as biological indicators and frequently employed. Other heat-resistant spore-forming microorganisms such as Clostridium sporogenes, Bacillus subtilis, and Bacillus coagulans have also been used in the development and validation of moist heat sterilization processes.
For dry heat sterilization, spores of Bacillus subtilis
spp. are sometimes used to validate the process. During the validation of dry heat sterilization processes, endotoxin depyrogenation studies are frequently conducted in lieu of microbial inactivation studies during the establishment of sterilization cycles because the inactivation rate of endotoxin is slower than the inactivation rate of Bacillus subtilis
spores. In practice the reduction of endotoxin titer by three or more logs will result in a process that also achieves a probability of nonsterility substantially lower than 106
Spores of Bacillus pumilus have been used to monitor sterilization processes using ionizing radiation; however, this is a cedlining practice. Radiation dose-setting methods that do not use biological indicators have been widely used to establish radiation processes. Furthermore, certain bioburden microorganisms can exhibit greater resistance to radiation than Bacillus pumilus.
For ethylene oxide sterilization, spores of a subspecies of Bacillus subtilis (Bacillus subtilis var. niger) are commonly used. The same biological indicator systems are generally used when 100% ethylene oxide or different ethylene oxide and carrier gas systems are used as sterilants.
Vapor-Phase Hydrogen Peroxide (VPHP)
This process has been shown to be an effective surface sterilant or decontaminant. VPHP is capable of achieving sterilization (probability of nonsterility of less than one in a million) when process conditions so dictate and if the target of sterilization is suitably configured. However, VPHP is also commonly used as a surface decontaminating agent in the treatment of sterility testing, biological and chemical containment, manufacturing isolators, and clean rooms.
Surface decontamination is a process that is distinct from sterilization of product contact materials, container-closure systems, or product. It is a process designed to render an environment free of detectable or recoverable microorganisms. Biological indicators are widely used to verify the efficacy of the decontamination process. However, in the case of decontamination, a spore log reduction value of three to four is adequate because the goal is decontamination rather than sterilization.
Bacillus stearothermophilus is the most prevalently used biological indicator for validating VPHP. Other microorganisms that may be useful as biological indicators in VPHP processes are spores of Bacillus subtilis and Clostridium sporogenes. Other microorganisms may be considered if their performance responses to VPHP are similar to those of the microorganisms cited above.
These spores may be inoculated on the surface of various gas-impervious carrier systems having glass, metal, or plastic surfaces. Highly absorbent surfaces, such as fibrous substrates, or any other substrate that readily absorbs VPHP or moisture may adversely influence the VPHP concentration available for inactivation of inoculated microorganisms. Paper substrates are not used because VPHP will degrade cellulose-based materials.
For representative characteristics of commercially supplied biological indicators, see Table 1.
Table 1. Typical Characteristics for Commercially Supplied Biological Indicator Systems
||Example of a Typical D value (minutes)
||Range of D values for Selecting a Suitable Biological Indicator
|Limits for a Suitable Resistance (depending on the particular D value [minutes])
|600 mg per liter
|60% relative humidity
For 1.0 × 106 to 5.0 × 106 spores per carrier.
For 1.0 × 106 to 5.0 × 107 spores per carrier.
For 1.0 × 105 to 5.0 × 106 spores per carrier.
The biological indicator may also be individually packaged in a suitable primary overwrap package that does not adversely affect the performance of the indicator, and is penetrable by VPHP. Spunbound polyolefin materials have proven to be well suited as an overwrap of biological indicators intended for use in evaluation of VPHP processes. The overwrap material may facilitate laboratory handling of the biological indicators following exposure to VPHP. Also, the use of an overwrap material to package VPHP biological indicators must be carefully assessed to ensure that, following VPHP exposure, residual hydrogen peroxide is not retained by the packaging material, possibly inducing bacteriostasis during the recovery steps. Microbial D values will be influenced by the presence of a biological indicator overwrap material relative to the rate of inactivation and the potential presence of residual VPHP. In cases where biological indicators (inoculated carriers) are being used without the primary package, stringent adherence to aseptic techniques is required.
USE FOR IN-PROCESS VALIDATION
Regardless of the mode of sterilization, the amount of the initial population of the microorganisms, its resistance to sterilization, and the site of inoculation on or in the product can all influence the rate of biological indicator inactivation.
During product microbial challenges, various areas of the product should be inoculated with biological indicators. If, for example, a container with a closure system is sterilized, both the product solution and the closure should be challenged to ensure that sterilization equivalent to a 106
(one in a million probability of a nonsterile unit) sterilization assurance level (SAL) will be obtained in the solution as well as at the closure site.
One may need to determine through laboratory studies whether product components are more difficult to sterilize than, for example, a solution or drug within the product. Depending on the locations of the product components most difficult to sterilize, different process parameters may be involved in assuring microbial inactivation to an SAL of 106
. The product performance qualification phase should identify the most important process parameters for inactivation of microorganisms at the sites most difficult to sterilize. Once these critical processing parameters are determined, during sterilization in-process validation of the product, they should be operated at conditions less than the conditions stated in the sterilization process specifications. Biological indicator survival is predicated upon both resistance and population. Therefore, a 106
biological indicator population is not always required to demonstrate a 106
SAL. The appropriate use for biological indicators is to employ them to confirm that the developed process parameters result in the desired SAL. In moist heat sterilization, the biological indicator is used to establish that physically measured lethality can be verified biologically. Biological indicators with substantive D values
and populations substantially less than 106
are adequate to validate many sterilization and decontamination processes. It is important that the users be able to scientifically justify their selection of a biological indicator.